- LiteratureApplication Notes Adsorbent Resins Guide Mass Spec Tips SDS Sheets FAQ MS Calibration Compound Spectra Manuals MS Links/Labs/ Organizations MS Online Tools Flyers on Products/Services Scientific Supplies Catalog About Us NextAdvance Bullet Blender® Homogenizer Protocols Micro-Mesh® Literature Instrumentation Literature Agilent GC/MS Literature SIS News / E-Mail Newsletter NIST MS Database - Update Notifications
- Mass Spec Tips1 - Freon for use in Mass Spectrometer Leak Checking 2 - Mass Spectrometer Probe Cooling 3 - Sample Vials for Direct Probes 4 - Selection of Vacuum Pump Oils for Lowest Mass Spec Background 5 - Determination Linkages in Biomolecules of Disulfide 6 - Transfer of H.P. ChemStation MS and GC Chromatograms from PC to MAC Computer 7 - Repairing Clogged Thermospray Probes 8 - Tuning a Finnigan 5100 to Meet BFB or DFTPP Criteria 9 - H.P. 5971 Transfer Line Tip for Direct Introduction of Capillary Column 10 - Troubleshooting Finnigan 5100 GC/MS Systems 11 - Leak Checking Mass Spectrometers 12 - Elimination of Memory Peaks and GC Background Noise 13 - SuperIncos Mapped Software Print Buffer Lockup 14 - Reduction of Peak Tailing 15 - Electron Multiplier Sensitivity 16 - INCOS Procedure for Calibrating on the Finnigan 4500 17 - Extending Electron Multiplier Life 18 - What techniques or methods do you use to determine if the electron multiplier 19- What techniques or methods do you use to detect vacuum leaks in your mass spectrometer 20 - Extending Lenear Range of the Mass Spec Article - Improving Sensitivity in the HP 5971 Mass Spectrometer - Part 1 and Part 2
- MSAgilent Bruker Extrel JEOL Kratos Perkin Elmer SCIEX Shimadzu Thermo Varian Waters (Micromass/VG) FLIR/Griffin Inficon/Leybold/Balzers MKS/UTI Hiden Dupont/CEC Nermag Vestec Filaments Heaters/Sensors Wire Material Electron Multipliers Probe/Sample Vials Ion Transfer Tubes Calibration Compounds MALDI-TOF Supplies Ceramic Insulators Cleaning Supplies Other Filament Repair Source Cleaning Transfer Line Repair GC Injection Port Repair Vacuum Feedthru Repair Other Repair Services NIST MS Library Wiley MS Libraries SIMION® 8.1 Catalog Page A1 Catalog Page B1
- 9 - H.P. 5971 Transfer Line Tip for Direct Introduction of Capillary Column (This Page)
Author: John J. Manura
Affiliation: Scientific Instrument Services
In the original gold plated transfer line tip on the H.P. 5971 MSD (H.P. Part # 05971-20305), the GC capillary column is purged inside this fitting and diverted through side vents and then vented into the MSD source. This original H.P. design exposes the analytes being chromatographed to a hot metal surface before they enter the MSD source. Loss of sensitivity for some compounds can result from both the indirect transfer line path and also due to sample decomposition upon exposure of thermally labile compounds to the hot metal surfaces.
As a result we have redesigned the Transfer Line Tip for the H.P. 5971 MSD to permit the end of the fused silica capillary column to purge directly into the MSD source. The new S.I.S. design eliminates any metal contact with the sample - the GC capillary column protrudes directly through the Transfer Line Tip and into the MSD source. This new design should result in higher sensitivity due to this straight path for sample flow and the elimination of contact by the compounds being chromatographed to any metal surface in the transfer line.
When inserting the capillary column through the MSD interface and into the MSD source, it is necessary to remove the MSD interface in order to properly position the capillary column at the end of the Transfer Line Tip, since there is no physical stop at the end of this fitting. After the capillary column with accompanying nut and ferrule are attached to the MSD interface, the capillary column should be adjusted so that the capillary column protrudes past the end of the Transfer Line Tip between 0 and 1.0 mm. After this distance has been set the GC nut and ferrule should be tightened to seal the capillary column in the correct position and the MSD interface can be reattached to the MSD.
The new Transfer Line Tip is available from Scientific Instrument Services.
(Part # HP99, price $79.00 each.)