- LiteratureApplication Notes Adsorbent Resins Guide Mass Spec Tips SDS Sheets FAQ MS Calibration Compound Spectra Manuals MS Links/Labs/ Organizations MS Online Tools Flyers on Products/Services Scientific Supplies Catalog About Us NextAdvance Bullet Blender® Homogenizer Protocols Micro-Mesh® Literature Instrumentation Literature Agilent GC/MS Literature SIS News / E-Mail Newsletter NIST MS Database - Update Notifications
- MSAgilent Bruker Extrel JEOL Kratos Perkin Elmer SCIEX Shimadzu Thermo Varian Waters (Micromass/VG) FLIR/Griffin Inficon/Leybold/Balzers MKS/UTI Hiden Dupont/CEC Nermag Vestec Filaments Heaters/Sensors Wire Material Electron Multipliers Probe/Sample Vials Ion Transfer Tubes Calibration Compounds MALDI-TOF Supplies Ceramic Insulators Cleaning Supplies Other Filament Repair Source Cleaning Transfer Line Repair GC Injection Port Repair Vacuum Feedthru Repair Other Repair Services NIST MS Library Wiley MS Libraries SIMION® 8.1 Catalog Page A1 Catalog Page B1
- MS Online ToolsIsotope Distribution Calculator and Mass Spec Plotter Mass Spectrum Generator Exact Mass Calculator, Single Isotope Version SIS Online GC Chromatogram and Mass Spectrum Viewer Search NIST 17 and Wiley MS Databases Mass Spec Calibration Compounds and Spectra Exact Masses of the Elements and Isotopic Abundances
- Mass Spectrum Generator (This Page)
Directions: Key in your mass spec peaks and relative abundances in the "Ion Masses / Intensities" box below. Each peak should be entered as a separate line. Each line should contain a mass field, followed by whitespace, followed by an intensity field.
The mass range is 1 to 100,000. Abundances should be expressed in percents of the base peak, or select the "Normalize MS Peaks" check box to normalize all mass intensities relative to the most abundant peak.
Developed by John J. Manura, Scientific Instrument Services,
copyright 1996-2016 Scientific Instrument Services.
Scientific Instrument Services is not responsible for any errors which may result from the use of this program.