- LiteratureApplication Notes Adsorbent Resins Guide Mass Spec Tips SDS Sheets FAQ MS Calibration Compound Spectra Manuals MS Links/Labs/ Organizations MS Online Tools Flyers on Products/Services Scientific Supplies Catalog About Us NextAdvance Bullet Blender® Homogenizer Protocols Micro-Mesh® Literature Instrumentation Literature Agilent GC/MS Literature SIS News / E-Mail Newsletter NIST MS Database - Update Notifications
- Mass Spec Tips1 - Freon for use in Mass Spectrometer Leak Checking 2 - Mass Spectrometer Probe Cooling 3 - Sample Vials for Direct Probes 4 - Selection of Vacuum Pump Oils for Lowest Mass Spec Background 5 - Determination Linkages in Biomolecules of Disulfide 6 - Transfer of H.P. ChemStation MS and GC Chromatograms from PC to MAC Computer 7 - Repairing Clogged Thermospray Probes 8 - Tuning a Finnigan 5100 to Meet BFB or DFTPP Criteria 9 - H.P. 5971 Transfer Line Tip for Direct Introduction of Capillary Column 10 - Troubleshooting Finnigan 5100 GC/MS Systems 11 - Leak Checking Mass Spectrometers 12 - Elimination of Memory Peaks and GC Background Noise 13 - SuperIncos Mapped Software Print Buffer Lockup 14 - Reduction of Peak Tailing 15 - Electron Multiplier Sensitivity 16 - INCOS Procedure for Calibrating on the Finnigan 4500 17 - Extending Electron Multiplier Life 18 - What techniques or methods do you use to determine if the electron multiplier 19- What techniques or methods do you use to detect vacuum leaks in your mass spectrometer 20 - Extending Lenear Range of the Mass Spec Article - Improving Sensitivity in the HP 5971 Mass Spectrometer - Part 1 and Part 2
- 13 - SuperIncos Mapped Software Print Buffer Lockup (This Page)
Author: Dennis Beauchamp
Affiliation: Weston Analytics
If your Finnigan system has dual terminal operation chances are you have Mapped software. This software is a big improvement over the older revisions but it does have a few quirks. The one that I have had the most is that the printer stops printing. Turning the printer on and off and Rebooting the computer won't fix the problem but the following will.
Enter IDOS (+)
In IDOS (+) type BOOT 0:
This will put your system into the unmapped environment. Remaining in IDOS (+), delete the offending files using the DE command.
The files you will most likely be deleting are:
The easiest way to delete these files is to type the following:
+DE 1:PPQ*.* using the *wild cards will delete all PPQ files.
+DE 1:PPRTPLT*.* again using the *wild cards will delete all PRTPLT files.
After all the files have been deleted then you will need to boot back into the mapped environment.
in IDOS (+) type BOOT 1:
You now should be able to print again. When you first try to print after deleting the files there may be a delay before the printer starts to print. Be patient it sometimes takes a little time to get the printer started again.
I have seen this problem on 5100's, 4500's and Incos 50's.